CR10X measures on-board CO2 levels and temperature
South Africa's subtropical climate is favorable for citrus fruit production. South Africa annually exports roughly one million tons of citrus fruit, comprised primarily of oranges, grapefruit, and lemons, to various overseas destinations. During shipment, the carbon dioxide (CO2) and temperature levels surrounding the fruit play a vital part in preserving its freshness. A typical shipment from Maputo, South Africa to the USA takes about two weeks. During transit, many shipping companies promise to keep the CO2 and temperature within acceptable tolerances.
To confirm this, the South African Citrus Research Institute initiated a project to measure CO2 and temperature levels during a routine shipment to see whether the prescribed levels are continuously maintained. To reduce the cost of the study, the challenge was to measure CO2 levels at eight points within the citrus cargo hull without using eight CO2 analyzers. CS Africa developed the monitoring system using a CR10X datalogger, a VICI8-port rotary valve (stepper motor controlled), a LI-COR LI-820 CO2 analyzer, hundreds of meters of tubing, and a vacuum pump.
A standard measurement cycle consisted of setting the rotary valve to the first port with a CR10X control port, initiating the vacuum pump, drawing a sample for 60 seconds through the connected tube, taking a CO2 reading with the LI-820, then stepping the valve to the next position with a CR10X control port. This sequence was repeated until all eight lines were measured. Temperature was also simultaneously logged using thermocouples.
To accommodate the client's limited budget, CS Africa could not use an expensive vacuum pump; instead, we effectively modified an inexpensive air pump. The first instrumented shipment showed the measurement system worked as planned for the two week travel time. Beginning on day two, the CO2 level not only exceeded the prescribed limits, they exceeded the LI-820's measurement range of 5000 ppm. To determine what the CO2 levels are throughout the shipment, a 20,000 ppm optical bench has been fitted for the next trip.
- Future arrives with the new CR1000
- CR1000: A new generation, a landmark product
- CSI a natural for remote wind assessment systems
- Adjustable mounting platforms highlight new, lightweight tripods
- CR1000 enhanced by PAKBUS communications
- Programming flexible with CR1000
- CS Africa: Sub-Saharan affiliate offers sales, technical support
- PAKBUS® features and the CR1000's capabilities
- CS100 measures barometric pressure
- TX312 GOES Satellite Transmitter available
- South African Citrus Research Institute monitors exports
- Availability, service will continue for CR10X customers
- Take advantage of the power of PAKBUS and features of the CR1000
- CR1000 easily added to your network
- CR1000KD: Take it everywhere